UNI 9213 (Parte 1a – 6a ), IWTO 33-34, ASTM D1909, UNI 8476 - ISO 3074, ASTM D2495-07
|Regain tester 70|
|Regain tester 70 + HMI touch|
|Manutenzione - Servicing|
|Stampante - ASCII printer|
|Software - Software for analyses|
The air, suitably filtered with an air inlet pre-filter, is heated up by power heaters commanded with a special close-control able to maintain and to keep stable temperature under the closest tolerances at +/- 0,5°C.
During the preparation the samples have to be put inside dedicated stainless steel baskets (equivalent to the "Tare"); after the initial weighing operations for tare and original wet mass determination, the operator can start the drying set at the SetPoint Temperature up the end of process determining the dry mass of material.
An HMI User interface with Touch Screen TFT 3" allows the operator to manage and monitor easily every single step across setting and manual operations described on the screen. Moreover all test results are stored on a saved memory or released from an ASCII printer in paper format mounted on board or transferred on and Excel Workfile with external application software.
The manual regain tester 70 has to be powered from a 3-phases power supply of 400VAC/50Hz with neutral (N) and protective conductor (PE). Different power supply information has to be noticed to our technical department before the preparation of the product.
|Power supply:||400 VAC/50 Hz + N+ PE|
|Temperature:||50-105 °C (on request up to 140 °C)|
|Temperature accuracy:||± 0,1 °C|
|Weighing range:||0..6100 g|
|Scale reading accuracy:||+/- 0,01 g|
|Basket diameter:||245 mm|
|Basket height:||310 mm; 190 mm;|
|Basket standard mass:||1600 g at least|
|Test time:||< 5 minutes on fibers;|
|<10 minutes on yarn bobbins|
|Power 1° heaters group:||5 kW|
|Power 2° heaters group:||5 kW|
|Main blower power consumption:||0,7 kW|
|Maximum power installed:||11 kW (400 V)|
|External dimensions: (L x P x A):||1030 x 630 x 1930 mm|
|Net weight:||170 kg|